Washington, DC: Abbott is recalling FreeStyle lnsulinx Blood Glucose Meters in the United States, due to issues with incorrect test results. At extremely high blood glucose levels of 1024 mg/dL and above, the FreeStyle InsuLinx Blood Glucose Meter will display and store in memory an incorrect test result that is 1024 mg/dL below the measured result.
The FreeStyle InsuLinx Blood Glucose Meter measures sugar (glucose) in blood drawn from the fingertips of people with diabetes to monitor blood sugar levels. On April 15, 2013, Abbott Diabetes Care sent an Urgent Product Recall letter to all its affected customers. The FreeStyle InsuLinx Blood Glucose Meters were distributed from April 18, 2012 through April 1, 2013.